Keithley 428 configuration¶
Keithley 428 is a amplificator that converts fast, small currents to a voltage, which can be easily digitized or displayed by an oscilloscope, waveform analyzer, or data acquisition system.
It cannot measure a current, just change the gain of the amplifier which can be read via a V2F and a P201.
Yaml sample configuration¶
- class: keithley428 plugin: bliss name: k_diode1 gpib: url: enet://gpibid42 pad: 22
- Read-Write attributes:
voltage_bias: Voltage Bias (-5V ; +5V)
current_suppress: Current suppress
*Read Only attributes:
Shell info example:
DEMO : k1 Out : KEITHLEY K428 COMM: GPIB type=ENET url='enet://gpibid42a.esrf.fr' primary address='11' secondary address='0' tmo='13' timeout(s)='0.5' eol='' gain: (3, '1E03V/A') filter_rise_time: (0, '10usec') voltage_bias: 0.0 state: K428 - Display:Normal - VBias:off - Zcheck:on - gain:1e03 - rise time:10usec overloaded: False filter state: Off auto_filter_state: On zero_check: On